Sub MkIIIF for diving
The Sub MkIIIF is a hyperbaric atmosphere monitoring system ideal for use with decompression chambers.
Can Uzunsoy of Istanbul Shipyard says: "Your (Sub MKIIIF) analysers are very important for our ships (rescue and underwater systems abilities), you are a partner of valuable efforts"
The Sub MkIIIF consists of a central display which can be connected to up to 3 remote sensor modules or ‘REM’s’.
The DCM would sit in the chamber control panel, and the REMs would be located inside the chamber.
A single lock chamber would typically require 1 x REM 1 plus 1 x REM 2
A twin lock chamber would typically require 2 x REM 1 plus 1 x REM 2
A maximum of 3 REMs could be connected to 1 DCM.
The Sub MKIIIF is controlled by an operator console known as the ‘display control module’, or ‘DCM’. This displays measured parameters and alarm states sent from the remote sensor modules (REM) which are connected to it.
A REM 1 is designed to measure partial pressure oxygen, partial pressure carbon dioxide, and pressure
A REM 2 is designed to measure temperature and humidity
If the measured parameters go outside normal operating limits there are two levels of audio-visual alarm activated, on both the DCM and the REM units. These alarms are announced by audible sounders and visible alerts (indicator lamps) on the front panel of each unit, which provide warning of gas and environment alarms, communication faults and calibration/general faults.
The DCM offers datalogging capability and can provide both relay outputs and 4–20mA outputs. The DCM is powered using 24 V DC. The REMs can be powered from either the DCM or a local supply. The DCM and REM units are connected together via a communications cable.
The REM units are designed to be calibrated and maintained by the user. A calibration check is recommended prior to pressurisation, or once every three months.
Spares and accessories
|MIRBGNX4S||0 to 50mBar CO2 sensor in N2|
|SA1HAT690||Spare temperature and humidity sensor for REM2|
Spare cable DCM to REM1 or REM2
Spare cable from REM2 to REM1
|FSUB-609||Spare cable DCM to terminal|
|2822-0019A||Universal in-line power adaptor 24V 65W|
|2822-0023||DIN rail mount power adaptor|